THE C-MOUNT, A NEEDFUL THING
For beginners, the combination of a light microscope with a digital camera includes some surprises. There are obvious discrepancies between the visual impression in the eyepieces and on the screen respectively. These discrepancies are about colour impression and image section.
The colour impression in the eyepieces may be taken as a reference and depends on the colour correction of the objective. The Plan Apochromatic lens design gives best preconditions for maximal colour fidelity and resolution. The imperfection of a secondary spectrum, caused by the usage of white light, is compensated for 3 colours, while spherical aberrations are corrected for 2 wavelengths. The result (1) is an impressive image quality without any coloured fringes. Especially in colourless black/white samples the absence of secondary spectrum becomes obvious. However, for price reasons, it is the Plan Achromatic quality level (2) of a modern Infinity System which sets the point of reference for most users.
Many beginners expect a 100% colour transfer to the screen via digital camera. But the sensory hardware of the digital camera, a CMOS or CCD chip, acts as a bottleneck. CCD targets are superior to CMOS sensors in terms of colour reproduction, but they still lack the dynamic range of the human eye. If it comes to the price point, a CMOS sensor will find his friends. A clever signal processing will lead to a reasonable image quality at an affordable price point.
There is one more problem to be solved, as the circular intermediate image within the microscope has to be captured by a rectangular sensor. Microscope manufacturers tend to enlarge the visible area (diameter of the intermediate image = Field Of View; FOV). Chip manufacturers go the opposite direction as the consumer market for pocket cameras and mobile phones requires small-sized sensors.
The combination of a typical microscope camera with the photo port of a microscope requires an intermediate piece of hardware, combining optical and mechanical tasks : the C-mount adapter (3).
Its upper part carries a standardized thread of 1 inch outer diameter and 1/32 inch pitch, ready for the mounting of the camera (4).
Its lower part refers to to the microscope manufacturer and the respective microscope model (5). Despite some homogenisations, even within one brand there might be different interfaces for different microscopes.
A C-mount 1X without lens is a pure mechanical device for a direct positioning of the camera sensor “on top” of the intermediate image (6).
This combination gives a disappointing image section of 10% Field of View (7).
A camera adapter with a magnification factor ≤ 1 will help. The adapter’s optics “compresses” the intermediate image and thus increases the captured area (8).
Feel free to reach out with our contact form here or chat with us directly if you have any immediate questions on our products and their applications.